|recovered_host||nutnr_b_dcl_conc_instrument_recovered||Data Products||Report M2M Stats||Science|
|recovered_host||nutnr_b_dcl_dark_conc_instrument_recovered||Dark Sample Data Products||Report M2M Stats||Science|
|recovered_host||nutnr_b_dcl_dark_full_instrument_recovered||Dark Sample Data Products||Report M2M Stats||Science|
|recovered_host||nutnr_b_dcl_full_instrument_recovered||Data Products||Report M2M Stats||Science|
|recovered_inst||nutnr_b_dark_instrument_recovered||Dark Sample Data Products||Report M2M Stats||Science|
|recovered_inst||nutnr_b_instrument_recovered||Data Products||Report M2M Stats||Science|
|telemetered||nutnr_b_dcl_conc_instrument||Data Products||Report M2M Stats||Science|
|telemetered||nutnr_b_dcl_dark_conc_instrument||Dark Sample Data Products||Report M2M Stats||Science|
|telemetered||nutnr_b_dcl_dark_full_instrument||Dark Sample Data Products||Report M2M Stats||Science|
|telemetered||nutnr_b_dcl_full_instrument||Data Products||Report M2M Stats||Science|
|Deployment||Cruise||Start Date||Stop Date||Mooring Asset||Node Asset||Sensor Asset||Latitude||Longitude||Deployment Depth||Water Depth|
No notes yet.
|Metadata||Start Date||End Date||Comment|
||9/10/14, 2:50 PM||8/22/15, 7:00 AM||
Deployment 1: Recovered_instrument data are not available for this deployment.
Id: 1502 By: lgarzio
||2/14/15, 7:00 PM||8/21/15, 8:00 PM||
Deployment 1: Status data sent from the buoy included leak detects in the buoy well, drop in battery voltage, and loss of wind turbine input. Upon recovery, the buoy was primarily intact but several instruments were damaged and/or missing. Ice build-up on the tower is speculated to be the cause for much of the damage. No telemetered or recovered_host data expected. Functional instruments could continue to collect data using internal battery power and storage cards.
Id: 46 By: lgarzio
||1/26/16, 7:00 PM||7/18/16, 8:00 PM||
Deployment 2: A period of violent weather caused power outages on multiple instruments. No telemetered or recovered_host data expected. Functional instruments could continue to collect data using internal battery power and storage cards. Upon recovery, the buoy well was flooded.
Id: 47 By: lgarzio
||12/26/16, 7:00 PM||4/6/17, 8:00 PM||
Deployment 3: Wind turbines were disabled due to evidence of ice build-up on the surface buoy tower. The platform's battery charge state declined steadily. As of 2017-01-15, all DCLs were shut down and the telemetry schedule was reduced to conserve power. Battery-powered instruments should continue to collect data. UPDATE 2017-04-07: With increasing daylight at this location, sufficient power was generated by solar panels to re-establish limited high-bandwidth telemetry and data transmission. UPDATE: at recovery, wind turbines were missing their hubs and blades, and one solar panel was missing.
Id: 121 By: lgarzio
||8/5/17, 2:17 PM||6/14/18, 4:02 AM||
Deployment 4: at 10:03 UTC on 12 October 2017 the Irminger Sea Surface Mooring stopped all communications. At recovery, the surface buoy was missing. The subsurface portion of the mooring had fallen to the seafloor and was recovered. CTDMOs, CTDBP-F, DOSTA, and ADCP instrumentation were full ocean depth rated and survived, although any recovered data available after 2017-10-12T10:03:00 are suspect. All other instrumentation imploded or flooded. The cause for the loss of the buoy is inconclusive.
Id: 102 By: lgarzio
||10/12/17, 5:00 AM||
The Satlantic ISUS instrument has been discontinued, and all OOI ISUS units have been converted to the Sea-Bird SUNA model. A new data parser is in development, and any resulting data gaps will be filled once the parser has been delivered and the data are processed.
Id: 1417 By: lgarzio