|recovered_host||dosta_abcdjm_ctdbp_p_dcl_instrument_recovered||Data Products||Report M2M Stats||Science|
|telemetered||dosta_abcdjm_ctdbp_p_dcl_instrument||Data Products||Report M2M Stats||Science|
|Deployment||Cruise||Start Date||Stop Date||Mooring Asset||Node Asset||Sensor Asset||Latitude||Longitude||Deployment Depth||Water Depth|
No notes yet.
|Metadata||Start Date||End Date||Comment|
||3/15/15, 5:25 PM||11/26/15, 5:08 AM||
Deployment 1: Buoy log files showed that seawater was first detected inside the buoy well on 3/26/2015. Both leak detectors showed increasing presence of water over the period of deployment. Upon recovery of the mooring, no external damage was visible and the buoy did not appear to be floating lower than normal.
Id: 653 By: lgarzio
||11/14/15, 4:06 PM||
Dissolved Oxygen concentrations are currently not being calculated due to a function error. Issue under investigation (Redmine 13228).
Id: 783 By: lgarzio
||10/26/16, 8:00 PM||
The derived O2 products (L1 Temperature Compensated and L2 Salinity Corrected Dissolved Oxygen) are suspect, due to errors during vendor calibration. Until a correction can be applied, they should not be used from 10/27/2016 onwards.
Id: 86 By: lgarzio
||10/26/16, 9:47 PM||5/11/17, 4:41 PM||
Deployment 3: Upon recovery, the cable to the CTDBP was severed.
Id: 389 By: lgarzio